Mais la polymyxine n'est pas du tout absorbée dans le sang du système gastro-intestinal et n'a d'effet que dans l'intestin et est utile pour le traitement des infections intestinales doxycycline prix Internet en y faisant des achats permettant d’économiser jusqu'à soixante-dix pour cent, tout en étant sûr de la qualité des produits pharmaceutiques.
Warnings and precautions
Sample collection and preparation
SensoLite Nova Test SensoLite Nova Test
blood glucose strips are for Preparation
IN VITRO diagnostic use only (external use only).
SensoLite Nova Test
blood glucose strips are designed
Blood Glucose Test Strip
Do not use test strips after their expiration date.
specifically for use with fresh capillary whole blood taken from
Do not use test strips that are wet, bent, scratched,
fingertip or earlobe. Venous blood, plasma, or serum samples
cannot be used. Blood testing must be performed immediately
Do not re-use the strips.
after the sample is obtained. Common anticoagulants and
Test strips for quantitative measurement of glucose concentration
If you have symptoms which are not consistent with your
preservatives such as heparine and sodium EDTA are allowed
in whole blood using with the SensoLite Nova
blood test results always contact your health care professional
to be used, but fluoride preservatives should be avoided.
SensoLite Nova Plus
blood glucose meter.
Testing your blood
Before you begin
If your reading result is above or below your expected
Refer to information described in chapter „Measuring with the
Carefully read this entire guide. If you have any question
range of blood glucose values, always repeat the test
meter” of the Instructions for Use of the SensoLite Nova
and/or need assistance, please, contact our authorised
and prove that the values are consistent.
SensoLite Nova Plus
Abnormal red blood cell quantity (hematocrit below 30% or
above 55%) may cause incorrect results.
Used strip and lancet are contaminated with
Never make significant changes in your diabetes treatment
blood. Be careful, follow the relevant, local
blood glucose test strip was designed to
program or ignore symptoms without consulting your
instructions when you throw the used strip and
provide an easy, accurate method for the determination of
capillary whole blood glucose content. This analysis applies
You must calibrate (code) the meter before the first use of
the enzyme glucose oxidase and is based on advanced
the system or when a new box of test strips is to be used.
electromechanical technology that is specific for ß-D-glucose
It is recommended to run the quality control test in (see
and SensoLite Nova Plus
measurement. Test strips are designed in such a way that
SensoLite Nova and SensoLite Nova Plus systems use
blood sample absorbs into the reaction zone, after blood has
internal coding system. A Code-card, being included in
the test strip package, holds encoded alfanumerical infor-
When your test results do not agree with your expectation
been applied to the tip of test strip. In the reagent zone glucose
mation that matches the Code value being affixed to
oxidase triggers the oxidation of glucose in blood. Intensity of
the bottle of test strips. Setting the code in the meter
When the test strips have been exposed to temperature,
formed electrons is measured by the meter and correlates
means entering the same code value into SensoLite Nova
not conforming to prescribed storage conditions (8°C to
well with the concentration of glucose in the blood sample.
glucose meter that is written on the strips’ bottle
label whenever you open a new bottle of test strips
When you open a new bottle of test strips.
10, 25 or 50 strips in a vial
(either by using the Code-card, or manually). The code
If the meter has been dropped.
value entered into the meter should match the code value
At least once per week to verify that the meter and test
Instructions for use
indicated on the strip bottle’s label and being encoded on
strips are working properly together.
Code-card. On how to enter the code number refer to the
When using Control Solution for system checking apply only
Measurement range of SensoLite Nova Test
chapter „Strip code setting” in the Instructions for Use of
CareSens Control Solution
, designed specifically for this
Sensolite Nova and Sensolite Nova Plus glucose meters.
system. CareSens Control Solution
can be obtained from
Manufacturer upon user’s request for additional charge (see
Storage and Handling
exact address and phone number of manufacturer at the end
One blood drop is enough to determine blood glucose value with
SensoLite Nova Test
blood glucose strips are packaged in
of this manual). Other control fluids provide incorrect results,
SensoLite Nova Test
strip. Minimum sample volume is 0.5 μl.
bottles. Store test strips at temperature between 8°C and
therefore cannot be used. When control solutions are used,
30°C (47°F-and 86°F) and between 30-70% relative humidity
test results should be within the expected range indicated on
Each test strip contains the following ingredients in the
and far from direct sunlight. Do not refrigerate or freeze. When
test strips bottle (L1, L2). Test results being within the given
stored properly, the unopened test strips are stable until the
range, indicate that the system (your meter and test strips) is
expiration date, indicated on the bottle. The month indicated,
working properly. If test results obtained with control solution
are not within the indicated range, always repeat the control
refers to the END of that month. Use the strips within
solution test. If the incorrect result still stands, contact your
3 months after first opening of their bottles. Never touch the
Each vial cap contains 2-3 g silica gel.
test strips with wet or contaminated hands.
Therapeutic levels of tetracycline may result inaccurate
Blood glucose test results are shown on the meter's LCD
display in either mg/dl or mmol/l measurement unit. Units can
Therapeutic levels of acetylsalicylic acid(e.g.: Aspirin) may
be set prior the tests (see chapter “Selecting measurement
result inaccurate(decreased) glucose readings.
unit” in Instructions for Use of SensoLite Nova
or SensoLite Nova
). The exchange rate of the units is: 1 mmol/l = 18 mg/dl.
If you get an unusual test result, always check the physical
Accuracy of SensoLite Nova Test
strip was assessed by
comparison of patients with diabetes in one tipical clinical
Check if the drop of blood is completely fills the reaction
evaluation. Reference measurements were done by another
approved blood glucose monitor. Specimens ranged from 1.0 mmol/l to 26.4 mmol/l, as measured by the reference method.
The linear regression data from one tipical clinical site studyis presented in the table below.
Check if the expiration date indicated on the test strips bottle
Check if the code number of the test strip in use matches
Check meter ability to function properly with the „Check
Test based on Standard EN ISO 15197:2003.
Check the system performance with CareSens Control
A laboratory study was conducted with SensoLite Nova Test
strip using fresh sodium EDTA venous blood with a broad
When test results are still questionable or inconsistent, and
range of glucose levels. Results from a single typical run of
your blood glucose results are less than 50 mg/dl (2.8 mmol/l)
this study, shown in the table, exhibit excellent precision
or greater than 300 mg/dl (16.7 mmol/l), consult your health
characteristics for SensoLite Nova Test
strip. From all the
care professional before making any changes in your
studies taken together, estimates of system precision were
calculated within-run precision, 3.65% between run precision,
• SensoLite Nova Test
strip is designed to be used with
fresh capillary whole blood samples. NEVER use serum
Hematocrit variation in sample: hematocrit between 30%
to 55% have no significant effect on test results. Very high
(above 55%) and very low (below 30%) hematocrits may cause inaccurate test results.
Neonates: Do not use SensoLite Nova Test
strip to test
neonates. The performance of SensoLite Nova
and SensoLite Nova Plus
system has not been validated with
Do not use fluoride as a preservative in blood samples.
N.Tietz, Fundamentals of Clinical Chemistry,
Abnormal blood specimens (i.e., high ascorbic acid, high
uric acid) may effect test results. Blood glucose readings
A.E.G. Cass, Biosensors: A practical approach,
in these cases should be interpreted for diagnoses with
D. Schade, „101 Tips for Improving your blood sugar", p 4-5.
FORMAS FARMACÊUTICAS: Xarope 0,4 mg/mL APRESENTAÇÕES: Embalagem com 1e 50 frascos de 120 mL + copo medida USO PEDIÁTRICO E ADULTO COMPOSIÇÃO: Cada ml contém: sulfato de salbutamol . 0,484 mg (equivalente a 0,4 mg de salbutamol) veículo q.s.p. . 1,0 mL (Veículo: sacarose, sacarina sódica, ciclamato de sódio, ácido cítrico, citrato de sódio, metilparabeno, propilparabeno,
Articles Risk-adapted craniospinal radiotherapy followed by high-dose chemotherapy and stem-cell rescue in children with newly diagnosed medulloblastoma (St Jude Medulloblastoma-96): long-term results from a prospective, multicentre trial Amar Gajjar, Murali Chintagumpala, David Ashley, Stewart Kellie, Larry E Kun, Thomas E Merchant, Shaio Woo, Greg Wheeler, Valerie Ahern, Matthew J